Please contact Lindsey Doyle (email@example.com) to be added to the e-mail list
Two days before data collection (i.e. if data collection is on a Thursday then crystals will ship out on Tuesday). Crystals must be loaded into the shipping dewar by 2:30 pm.
Hit the large, red EMERGENCY button on the generator’s front panel
Sign up on the white boards in the X-ray suite (B3-170).
Users have not been removing their data files. Check with other user to see if they can remove any files. Let Lindsey know so she can schedule a purge.
Using the PC in the X-ray enclosure, go to the AnaWin toolbar. Select ChannelSetup, and then Channel Parameters. Set channel1 (nozzle) to 4°C. Don't change channel2(pre-nozzle). Leave it at its preset. CHANGE IT BACK WHEN YOU ARE FINISHED!!!!
Call AirGas at 1-800-224-7427, press 2. Our customer number is 2336490.
Turn the helium compressor off overnight (Big Green button on Helium compressor, not the smaller control box)
Restart CrystalClear. If this doesn't work, restart the PC. If this doesn't work, shutdown both the PC and controller (blue box behind panel, under 4++). Leave controller off for ~15minutes, so capacitors lose charge. Restart.
Press the black “Silence Alarm” button on the control panel next to the Helium tanks on the back wall. Let Lindsey know.
That’s the cold-water alarm on the Haskris. Let Lindsey in B3-175 know (or Peter in B3-177 if Lindsey isn’t around).
Open the front swing door of the generator (below the control panel). Use a phillips head screwdriver to open the plexiglass window. Hit the reset button.
Pull out the reset button and turn it clockwise. Turn the power on. Once the run lamp is on and the vacuum is stable, turn on the X-rays and dial them up.